Alaris Signature 7130
The Alaris IVAC Signature Gold 7130 Infusion Pump is a single channel volumetric infusion pump that allows independent programming on both sides. It can be used for general IV infusions, antibiotics, blood infusions, enteral infusions, TPN and hyperalimentation, and lipids. Monitoring options provide increased awareness of occlusion detection and battery charge condition. Dualrate piggyback function and panel lock.
A multistep program function allows a sequential program to deliver up to nine rates and volumes; can also be programmed using volume and time. Multiple infusions can be preprogrammed over a period of up to 24 hours; delay start and loading dose features. Computer interface for remote monitoring. Rotating pole clamp for pole or bedrail mounting. Drugspecific dose rates can be calculated by selecting a drug name, which will then display on the screen. Recommended for epidural infusions. Optional flow sensor.
|Dimensions:||7.7” W x 8.6” H x 5.0” D (without power cord)|
|Power Cord:||10 ft|
|Case:||Impact and flame resistant plastic|
100120 V~,50/60 Hz, 0.5 A, 3wire grounded system (40 watts Max)
Rechargeable nickel cadmium. With a new, fully charged battery, the pump will operate for 4 hours nominal.
|Operating Conditions:||50-104° F and 2090% humidity, noncondensing|
|Storage Conditions:||-40 - 140° F and 595% humidity, noncondensing|
|Ground Current Leakage:||
Tested to UL 544 and CSA C22.2 No. 125 for medical and dental equipment.
|Mode of Operation:||Continuous|
|KVO Flow Range:||0.1 - 20.0 ml/hr in 0.1 ml/hr increments|
Flow Rate Range:
- 0.1 to 999.9 ml/hr in 0.1 ml/hr increments (primary)
- 0.1 to 270.0 ml/hr in 0.1 ml/hr increments (secondary)
|Flow Rate Accuracy:||
±5% at 1.0 999.9 ml/hr; ±6.5% at 0 1.0 ml/hr
|Volume To Be Infused Range:||
- 0.1 to 9999.9 ml in 0.1 ml increments (primary and dose rate modes)
- 0.1 to 999.9 ml in 0.1 ml increments (secondary)
|Volume Infused Range:||- 0.0 to 9999.9 ml in 0.1 ml increments|
|Alarms:||Accumulated Air in Line|
|Air in Line|
|Computer Link Failure|
|Flow Sensor Unplugged|
|Hold Time Exceeded|
|No Upstream Flow Detected|
|Primary Flow Detected during Secondary|
|Set Out |
|Set Up Time Exceeded|